E-Z Split Specimen Testing Information
The 8 Panel E-Z Split Key Cup Urine Drug Test detects the urine metabolites of the most common drugs of abuse within minutes. An additional feature is E-Z Split Key Cup’s unique activation key which preserves urine samples until you are ready to initiate the drug test.
- Amphetamines (AMP)
- Barbiturates (BAR)
- Benzodiazepines (BZO)
- Cocaine (COC)
- Marijuana (THC)
- Methamphetamine (mAMP)
- Opiates (OPI)
- Phencyclidine (PCP)
It's the most hygienic urine drug testing device on the market and is perfect for workplace use. Best yet, the flat results panel allows for photocopying of results for your records. A host of tamper protection features ensure the integrity of both sample and results.
For a quality drug test which is friendly to your budget, our 8 Panel E-Z Split Key Cup Urine Drug Test cup is built with your needs in mind.
E-Z Split Key Cup Instructions for Use:
Remove the key by twisting it from the center of the cup cap. Open cap with the Pull Tab.
Have the donor urinate directly into the cup and secure teh cap, making sure to provide a urine sample that's a minimum of 30ml. Note: Secure cap by pressing down on the Pull Tab until an audible click is heard.
Once the sample is returend to the collection official check the cap for a tight seal. Collection official dates and initials the security seal and places the security seal over cap.
On a flat surface, collection official inserts key and pushes in. Peel back the label to reveal the test panels and read the results at 5 minutes. Note: If the test cup includes Adulteration tests read results between 3-5 minutes. Results read after 10 minutes should be considered invalid
How to Read the Drug Test Results:
Negative drug test results will have two lines - one line in the control region and one line in the test region.
Presumptive positive results will show only one line in the control region. If a test fails to show a line in the control region, this indicates an invalid test result.
Invalid drug test results occur when no line appears in the control region. Should this occur, repeat the test with a new device. If results are still invalid, please stop using the test cups and contact a TestCountry representative.